I mistakenly left my digital multimeter on DC when i went to test a
wall outlet, and it threw a spark and charred the tips of my
multimeter. I had always been told it would not do such a thing, even
on the wrong setting. However, I am now timid to recheck the outlet
on the AC setting. So, I used a outlet plug light indicator and it
said everything was kosher with the outlet. Can anyone tell me why
response, but on DC setting one probe is for negative the other for positive
(highly simplistic). Pull the positive terminal off a car battery and
you'll see a spark too - basically the charge is jumping the gap until it
becomes too large to generate a potential difference. In your multimeter
when the AC switched from positive to negative the same thing probably
Like I said pretty simplistic answer, but it sounds right to me.
Or in DCmA,or ACmA.
If he had connected the 10A *unfused* range to 120VAC,it would have SHORTED
the outlet and popped a breaker,and made a REALLY big spark,and a loud
noise[KAPOW].It also would have melted the metal tip,perhaps also the probe
The shunt for the 10A range is generally a heavy piece of solid wire with a
tap for the output to the meter circuitry.
I believe you have to move the positive(red) lead to another jack to use
the 10A range on most DMMs.Some DMMs play it safe and you have to move the
red lead for ALL current ranges,for this reason.
You may want to check the internal mA fuse,it is probably open.
On 23 Feb 2007 16:33:13 -0800, email@example.com wrote:
It shouldn't spark at all. If the multimeter is of any decent design
it should have a high ohm/volt rating and should not present a
meaningful load to a circuit even on DC.
Did you rotate the switch while connected? Did you fumble one of
the leads? Did you have it on current?
On Feb 24, 6:52 am, firstname.lastname@example.org wrote:
I agree, in the DC mode, there shouldn't have been sparks. did the
meter have autorange? If you are concerned that the meter is mdamaged
and do not want a repeat spark event; try checking the meter against
itself in the ohms setting, or measuring a 1.5 battery, or a low
voltage AC source such as a no longer needed wall wart from an old
If the wall receptacle had a metal plate on it, you might have also
accidentally shorted a lead against it.
I suppose that if you do not use a meter on a regular basis, you would
purchase the least expensive one you could find, and you consequently
would also have minimal experience making measurements; if you use a
meter on a regular basis, you know that having a good quality meter is
imperative for safety and accuracy, and you also have expertise in
taking measurements. If you cooked your meter, and did not stop your
heart, then you paid a small price for a valuable lesson in taking
measurments safely, and selecting higher quality when shopping for a
hope this helps
As others indicated, if you were on DC
current, that is basically a short
circuit. Most, if not all meters have a
fuse inside to protect against just
that type of thing. If the fuse is now
open, the meter will still work on
AC and DC volts, and ohms scales, just
not on current settings. BTW,
if you use a multimeter a lot, changing
scales, etc., this is a pretty common
thing to happen.
This is not so. If such a circuit was in place every time you'd
measure DC you would have to have the polarity correct
otherwise you would blow this fuse you are talking about.
Fuses exist in some multimeters but they are not in the DC
voltage circuit. Instead a high impedance circuit is in place
whether it is AC or DC, + to - , - to +.
On Sun, 25 Feb 2007 13:28:20 -0500, email@example.com wrote:
I should read your post more carefully. I assumed you were referring
to DC voltage, not current.
Ignore the above reply as it is referring to voltage.
Dc current circuits do, as you say create a short that will spark
when used on voltage.
the person has to READ the side of the transformer/wall wart to
determine if it is DC or AC.gee whiz.
the suggestion was intended to provide Sparky with a safe low voltage
source to use to test the meter, that's all.
Hey Sparky, are you still there?
HomeOwnersHub.com is a website for homeowners and building and maintenance pros. It is not affiliated with any of the manufacturers or service providers discussed here.
All logos and trade names are the property of their respective owners.